Fetal Monitor can acquire fetal heart rate, maternal uterine contraction when pregnancies over 28 weeks to provide reference data for clinical use. The monitor can be used individually or connected with PC through RJ45 Interface for the purpose of central monitoring. It is only suitable for the equipment in hospitals, clinics, doctors offices and patients at home by professional medical personnel.
Light dexterous appearance, tops horizontally and walls can be hoisted
8.0 “screen color LCD display, rotatable screen to 60Â°
Display of the patient data and curve clearly
FHR 120 BPM~160 BPM normal range label
Manual records fetal movement
Sound and color alarm for high and low fetal heart rate
Continuous 24-hour real-time monitoring function
Continuous 12-hour patient curve and data storage, playback and print
Content: bed no. , pregnancy weeks, age, paper speed, date, time , volume, alarm status, transducer connection status, recorder status, FHR data and wave, Contraction data and wave, Fetal move times and mark etc.
Record Paper two-double type Z
Print Width: 112 mm
Valid Print Width: 104 mm
Paper output speed: 1 cm/minã€2 cm/minã€3 cm/min (optional)
Data Precision: Â±5 %ï¼ˆX Rollï¼‰ï¼ŒÂ±1 %ï¼ˆY Rollï¼‰
Record Content: hospital, bed No. ,name, pregnancy weeks, patient No., paper speed, date, time , FHR data and wave, Contraction data and wave, Fetal move times and mark etc.
Nominal Frequency: 1.0 MHz
Work Frequency: 1.0 MHzÂ±10 %
Negative peak sound pressure: p_ï¼œ1 MPa
Output beam intensity: Iobï¼œ20 mW/cm2
The peak time space peak intensity: Isptaï¼œ100 mW/cm2
The average time space peak intensity: Isataï¼œ10 mW/cm2
FHR Rang: 50 BPM~240 BPM
Resolution: 1 BPM
Accuracy: Â±2 BPM
TOCO range: 0~100 %
Resolution: 1 %
Nonlinear error: ï¼œÂ±10 %
RZ way: Manually
For the manual button (the operation of pregnant women), there will be a mark display in the bottom area of FHR wave display section.
Alarm for high and low FHR, which exceeds appointed limit.